The BLE radar sensor is used for non-contact level measurement in harsh environments and provides live data (every 10 seconds) with a battery life of up to 10 years. It convinces by its high measuring accuracy. This accuracy can be guaranteed even in narrow shafts thanks to a low radiation angle of 4°. To avoid parallax during installation, there is a spirit level on the BLE radar sensor.
This makes assembly even easier
The holder of the BLE radar sensor is made of stainless steel. The swivel arm allows the sensor to be aligned over the level to be measured. By means of a ball-and-socket joint, the alignment is subsequently made more precise. In addition, a laser distance meter can be attached directly to the sensor holder.
The 5 possibilities briefly explained
Mounting in the duct is done by means of a wall mounting flange (drill and screw).
The BLE radar sensor is mounted by means of a pipe mounting flange suitable for pipe diameters of min. 50 mm up to max. 275 mm.
The assembly is carried out by means of an adhesive flange. The substrate should be cleaned with a wire brush before bonding. 2K composite mortar injection material must be used as adhesive.
The sensor is mounted by means of a rung flange suitable for rung/bar diameters of min. 15 mm up to max. 26 mm.
A cross strut suitable for DN 600 manhole covers with pockets for dirt traps is used for installation.